A) Visual Examination
All The blades are checked by our well tarined staff.
Blades with defects like
nick, burr, wearivness, jag, bluntness are removed during the visual examination before the final
B) Hardness Testing
The hardness of the blade is maintained at a minimum of
750 HV for carbon steel blades and 700 HV stainless steel blades to achieve precise sharpness. The
hardness of the blade is checked at regular intervals on vickers hardness tester.
C) Checking of shapes, slot dimensions, Angle etc.
The shape and slot dimensions are checked on
shadowgraph in accordance with the standards specified.
D) Perfomance Test
100 mm long 5 cuts are made on chamols leather to check
the sharpness of the
blade.blades with uniform cutting edge which can cut the chamois leather cleanly without putting it,
are only passed
The blades are sterillised using the Gamma Radiation
process with a minimum does of 2.5 m rads.
All the blades are individually wrapped in aluminum
foll.100 such blades are packed in a carton. master packing can be changed as per moisture inside
Surgeon blades and medical devices (Services) Ltd uses latest's medical
technology to irradiate products in a state of the art computer controlled continuous irradiation
facility. The high quality service is available to other trader and suppliers to healthcare
we can provide a quality auditing service to determine the required absorbed dose to provide a
sterile medical device. The facility is routinely inspected by the FDA.
The company is certified to ISO 9001:2008
and ISO 13485:2003. [View our certificate here]
For further information or advice on technical aspects of the process please contact the company by
one of the following: [Contact us ]